Pulse Dose OxygenDose Delivery system Oxygen, known as intermitent or on demand oxygen what is it all about??
How does PulseDose work?
When inhaling, your diaphragm moves down and causes a drop in pressure in the lungs. Air flows in through the nose and mouth to equalize the pressure. This negative pressure is also present at the nose and mouth during inhalation. This pressure signal travels through the nasal cannula to a pressure sensor in the PulseDose conserving device. An electronic circuit then opens an electrical valve to deliver a precisely metered dose of oxygen. When the valve is closed, the sensor is ready to detect the next inhalation.
PulseDose dramatically extends the use time from a supply of oxygen, offering increased mobility with improved comfort and increased efficiency. The reliability and safety of PulseDose oxygen delivery has been proven effective in clinical testing.
Because oxygen is released only during inhalation, the constant flow of oxygen into the nostrils is eliminated. Many users find PulseDose oxygen delivery more comfortable than continuous flow delivery systems. The short ďpulseĒ of oxygen delivered during inhalation is almost undetectable, and the humidity in the room air helps maintain a normal level of moisture in the nasal cavity. This greatly reduces the discomfort of dehydration associated with a continuous flow oxygen system.
Am I really getting enough oxygen?
Yes. PulseDose delivers a precise burst of oxygen at a relatively high flow rate at the leading edge of each inhalation. This assures that the oxygen delivered flows deep into the lungs for maximum benefit. PulseDose requires less oxygen to deliver the same therapeutic benefit as continuous flow oxygen delivery.
I canít hear the pulse. Is PulseDose working?
If the pulse canít be heard, simply look at the green PulseDose indicator to see that the device is being triggered by inhalation. For further assurance, hold the end of the cannula in front of your lips while inhaling through your mouth and feel the pulse. PulseDose does not monitor the supply of oxygen. Remember to check the oxygen contents gauge periodically to verify that there is an adequate oxygen supply. If the oxygen supply runs out, the green PulseDose indicator light will continue to illuminate, indicating that the conserving device is being triggered by inhalation.
Why canít I use a cannula which is longer than 35 feet?
The PulseDose triggering is not significantly affected by the cannula length, but the delivery of oxygen is affected. If the cannula is longer than 35 feet, the pulse of oxygen is delayed. Remember the therapeutic moment during the inhalation cycle. If the oxygen is not delivered during this time, the benefits will not be realized.
Should I use a humidifier with PulseDose?
No. PulseDose is not able to sense inhalation through the water in the humidifier. Also, many patients find that humidification is not necessary with PulseDose. They find that PulseDose improves comfort because it delivers a very small amount of oxygen during the early part of inhalation, while the rest of the inhalation is composed of normal room air.
When Iím breathing faster, I donít get a pulse with each breath.
Donít I need a dose every time?
Because PulseDose breathes with the patient, it has an upper limit (40 Breaths Per Minute) that keeps you from getting too much oxygen. When breathing slowly, you receive a dose with every breath. As breath rate increases (up to 40 BPM) PulseDose still delivers a dose with every breath. At this point, you are getting more oxygen per minute because each pulse delivers the same amount of oxygen with each breath while the number of breaths has increased. With continuous flow oxygen, the oxygen delivered is constant. As you breathe faster, the enrichment of inhalations actually decreases because each breath is being diluted with a greater amount of room air.
Pulse dose was created to save and expand time on your oxygen tanks and liquid oxygen. When you have pulse dose on Oxygen Concentrators the pulse setting was created to get the concentrator to delivery higher settings on pulse which normally on constant flow it would not be able to do.
If you have any questions about the way all this works please feel free to call
Pulse only portable oxygen concentrators are the Inogen One G3, Respironics Evergo and Invacare XPO2.
Portable Oxygen Concentrator that will produce both pulse flow and constant flow is the Respironics SimplyGo, Sequal Eclipse and the Oxlife Independence.
Please call us at 1-877-901-1464 with any questions.
About American Medical